We are often asked about the difference between calibration, verification, and validation. While they may seem similar, it is critical that you use the correct process or processes to ensure that your quality system meets all the necessary requirements.
Calibration is the process of comparing an unknown quantity or value to a known quantity. A trained calibration technician will check the instrument at several points through its range and may make adjustments to bring the instrument into alignment with the standard if it is found to be out of tolerance.
All calibrations should be done using a traceable standard. Traceability means that the value of the standard can be directly traced back through an unbroken chain of comparisons to a national standard. These standards are coordinated by national metrological institutions, such as the National Institute of Standards and Technology (NIST).
Two levels of calibration that we offer at J.A. King are NIST Traceable and ISO 17025 Accredited. A NIST Traceable calibration is just that, a calibration that uses standards which can be traced back to NIST. An ISO 17025 calibration takes this a step further. When a laboratory says that they are an ISO 17025 Accredited laboratory, or that a calibration is ISO 17025 Accredited, it means that an accrediting body such as The American Association for Laboratory Accreditation (A2LA) has signed off on that lab’s procedures for calibrating specific equipment types within an approved range. All of the procedures that a lab has been signed off through an accrediting body are listed on their scope of accreditation. Take a look at J.A. King’s ISO 17025 scopes of accreditation.
Verification is a process that is done periodically to ensure an instrument is working as it should be and that any error in its measurement is still within tolerance for the application. Unlike calibration, this process does not involve making any type of adjustments. More importantly, verification may not require comparison against a NIST traceable master standard, it is simply a check against another piece of equipment to ensure the instrument is still working as intended.
For example, you may have a verification weight that you put on your balance each day to ensure it is reading correctly before using the balance. This isn’t a calibration, mainly because you aren’t checking the equipment over the entire range and because you aren’t making any adjustments to the equipment. Instead, you are simply ensuring that the instrument is working as intended.
Validation is particularly important in the life sciences industry, but has applications in other industries as well. Validation is the process of ensuring that all parts of a system are working together correctly to produce the desired outcome. A simple example is oven validation. You can calibrate and verify separate components such as temperature controllers, temperature monitors, and heating elements, but these components all work together in a system. This means you still need to confirm that the system functions correctly as a whole, ensuring that the oven is heating evenly throughout, at the desired temperature for the correct duration of time.
In any industry, specific procedures are written for validation protocols to ensure that the proper steps are taken and documented. In the life sciences industry, validation protocols are more formalized due to FDA requirements. These validation protocols know as IQ/OQ/PQ will include test parameters, product characteristics, production equipment, and decision points on what constitutes acceptable test results.
|What is IQ/OQ/PQ? Installation Qualification (IQ), Operational Qualification (OQ), and Performance Qualification (PQ) protocols are critical in equipment validation for quality assurance.|
The experts at J.A. King are here to help you with all of your calibration, verification, validation, and IQ/OQ/PQ needs. Contact us today to learn more about how we can assist you in improving your quality assurance process.